Interrogation Site Metadata
CZY - Crazyman Creek at 0.6 kmStatus: Active - Operational
General Operational Period: Seasonally
Start Date: 3/25/2014 12:00 AM
Instream detection system consisting of a tandum array located in the Imnaha River basin on Crazyman Creek. The array (CZY) is located at rkm 522.308.80.001 (N 45.22930 W -116.84478) and comprises of two arrays. Each array consists of two 10 foot Biomark light round black HPD antennas that spans the entire creek at base flow. Electrical power is from a thermal electric generator powered by eight 100 pound propane tanks. Remote communication is provided by a satellite phone modem. The site was installed on 11/8/2013.
|Last File Loaded:||CZY-2023-137-P-001.json|
|Date Last File Loaded:||5/26/2023 10:21 AM|
|Date Last File Opened:||5/17/2023 1:55 AM|
|Date Last File Closed:||5/25/2023 12:05 PM|
|Last Observation Tag:||3DD.003D495040|
|Last Observation Date:||5/22/2023 4:49 PM|
|Detection Summary:||Last Day's Observations|
Primary Contact:Cameron Albee
Data Steward:Neal Espinosa
Operational history describes the period the site was interrogating for PIT tags. The comments provide anecdotal information about the operation of the site during this period. Site operational periods are updated typically at the end of the calendar year.
|Year||Start Date||End Date||Comments|
|2022||3/15/2022 12:00 AM||6/17/2022 12:00 AM||Start and end dates based on timer tags.|
|2021||3/22/2021 12:00 AM||7/30/2021 12:00 AM||Start and end dates based on timer tags.|
|2020||4/10/2020 12:00 AM||7/27/2020 12:00 AM||Start and end dates based on timer tags. See event logs for additional operational information.|
|2019||4/2/2019 12:00 AM||7/26/2019 12:00 AM||Start and end dates based on timer tags.|
|2018||3/13/2018 12:00 AM||7/31/2018 12:00 AM||End date based on timer tags.|
|2017||3/10/2017 12:00 AM||6/25/2017 12:00 AM|
|2016||3/4/2016 12:00 AM||7/2/2016 12:00 AM|
|2015||3/27/2015 12:00 AM||8/8/2015 12:00 AM|
|2014||3/25/2014 12:00 AM||6/8/2014 12:00 AM|
Configuration metadata includes details about the equipment used at the site to detect and record PIT tags. A new configuration sequence record is created when equipment is added or removed from the site and the period of time it was active. A configuration sequence without an End Date is currently active.
|Antenna Group||Transceiver ID||Antenna ID|
Site Configuration Diagrams:
Select a type to view the history of equipment installed at this site. A missing End Date implies the information about the equipment is current.
|Equipment Code||Description||Start Date||End Date|
Event Log Summary
This displays the last 25 events, to see the full history, log in to the Event Logs.
|Event Date||Event Type||Comments|
|4/13/2023 12:00:00 AM||SiteUp||
Crazyman Creek Array (CZY) started collecting data April 13, 2023.
|6/16/2022 7:24:00 AM||SiteDown||
CZY site is down for the adult steelhead season.
|4/14/2022 10:00:00 AM||GMC||
Replaced FS1001M on 4/14/22 from 10:00 to 12:30. We were not seeing all the antenna test tags, now all 4 antennas are showing test tags in the CR1000 data logger.
|3/15/2022 2:50:00 PM||SiteUp||
Site CZY is up and running for adult steelhead season.
|5/15/2021 1:30:00 PM||SiteUp||TEG serviced and up and running.|
|5/14/2021 1:00:00 PM||SiteDown||Site offline to service TEG power source.|
|3/22/2021 10:00:00 AM||SiteUp||The site is up and running.|
|7/26/2020 7:26:00 PM||SiteDown||CZY is turned off for the year.|
|6/4/2020 11:30:00 AM||General||Test tags were not stored on the CR1000 data logger from FS1001M for some unknown reason. Duration 5/23/2020 to 6/4/2020.|
|5/9/2020 11:27:00 AM||General||Test tags were not stored on the CR1000 data logger from FS1001M for some unknown reason. Duration 4/10/2020 to 5/9/2020.|
|3/21/2020 3:15:00 PM||SiteUp||CZY array is up and running.|
|7/26/2019 1:00:00 PM||SiteDown||Powered down the TEG for the CZY array. CZY is off for the rest of the year.|
|4/17/2019 1:00:00 PM||General||Cambell scientific logger went down. Tag file was created from FS1001M tag file buffer. No test tags were recorded. Cambell scientific logger back up 4/30/2019 at 11:32.|
|4/17/2019 1:00:00 PM||General||No data was lost when Cambell scientific logger was offline.|
|4/2/2019 1:00:00 PM||SiteUp||CZY array is up and running at 13:19 on 4/2/2019.|
|3/13/2018 2:00:00 PM||SiteUp||CZY is up and running for the season|
|6/25/2017 5:40:00 PM||SiteDown||CZY ran out of propane and is off line for the year|
|3/10/2017 1:16:00 PM||SiteUp||Crazyman Creek PIT Tag array is up and running for 2017.|
|7/1/2016 10:50:00 AM||SiteDown||The Crazyman Creek PIT tag array ran out of propane to power the array from the TEG after 10:50 am on July 1, 2016. The Crayzyman Creek array is now offline for the year.|
|3/4/2016 1:00:00 PM||SiteUp||CZY site is up and running on 3/4/16 at 13:00|
|8/8/2015 12:00:00 AM||SiteDown||Ran out of propane. Site down for the year.|
|4/4/2015 9:00:00 AM||General||High noise at site. Replaced MUX and tuned antennas. Lowered noise.|
|4/3/2015 3:00:00 PM||General||Maintenance was performed on site. Moved gas line to ThermoElectric Generator. Site was down for 3 hrs (3:00 PM to 6:00 PM).|
|3/27/2015 3:00:00 PM||SiteUp||Fired up ThermoElectric Generator. Site is up and running for the year.|
|6/8/2014 8:10:00 AM||SiteDown||Crazyman PIT tag array ran out of propane and is off line for the season.|