Interrogation Site Metadata
Search by typing either a site name, site code, organization, or site type.
LMC - Lower Manastash CreekStatus: Active - Operations Unspecified
General Operational Period: Year Round
Start Date: 1/28/2014 12:00 AM
LMC is located in Manastash creek 1.4 miles above the confluence to the Yakima river, 80M above the bridge at Barnes Rd. All equipment was purchased by the Kittitas County Conservation District directly from Biomark. Site was installed and will be managed by WDFW personnel. Initial configuration of the site included 3-20ft, Biomark Lite antennas, each spanning the wetted width of the creek. Antenna 1 is upstream, 2 middle, and 3 is the downstream antenna. Antennas 2 and 3 were damaged February 2016 and were replaced in February 2017 with 24 x 8 ft quadrilaterlas constructed with 1 inch PEX. Ten inch posts were used on the bottom of the creek to create a backbone for the antennas. The posts are secured to the substrate with MR-4 anchors. The sides of the antennas are secured using anchors on an exposed bank, clamps on trees and one t-post for antenna 3. Antennas 2 and 3 have a complete field of detection with 24 inches of read range off the coil. Each antenna is connected to a submersible IS1001 antenna control node which are anchored to the bank. The exciter cable runs 120ft upstream along the left bank to the IS1001-MTS, secured in an enclosure. Grid power is being used to charge the 4-12V batteries located in a Jo-Box under the enclosure.
|Last File Loaded:||LMC-2021-111-P-ALL.xml|
|Date Last File Loaded:||6/14/2021 2:50 PM|
|Date Last File Opened:||4/21/2021 10:09 AM|
|Date Last File Closed:||6/14/2021 9:41 AM|
|Last Observation Tag:||3DD.003D45DBA7|
|Last Observation Date:||6/12/2021 11:48 AM|
|Detection Summary:||Last Day's Observations|
Primary Contact:Gabriel Temple
Data Steward:Zack Mays
Operational history describes the period the site was interrogating for PIT tags. The comments provide anecdotal information about the operation of the site during this period. Site operational periods are updated typically at the end of the calendar year.
|Year||Start Date||End Date||Comments|
|2020||1/1/2020 12:00 AM||5/13/2020 12:00 AM||Start and end dates based on timer tags. See event logs for additional operational information.|
|2019||1/1/2019 12:00 AM||11/14/2019 12:00 AM||Start and end dates based on timer tags.|
|2018||2/26/2018 12:00 AM||1/1/2019 12:00 AM||Start and end dates based on timer tags.|
|2017||3/1/2017 12:00 AM||6/23/2017 12:00 AM||Start and end dates based on timer tags. Reinstalled with replacement pass through antennas for 02 and 03 in February.|
|2016||1/1/2016 12:00 AM||5/3/2016 12:00 AM||Antennas 02 and 03 were damaged in February and site operated only with Antenna 01 after that.|
|2015||1/1/2015 12:00 AM||1/1/2016 12:00 AM|
|2014||1/28/2014 12:00 AM||1/1/2015 12:00 AM||Start and end dates based on timer tags.|
|2013||No data received.|
Configuration metadata includes details about the equipment used at the site to detect and record PIT tags. A new configuration sequence record is created when equipment is added or removed from the site and the period of time it was active. A configuration sequence without an End Date is currently active.
|Antenna Group||Transceiver ID||Antenna ID|
Select a type to view the history of equipment installed at this site. A missing End Date implies the information about the equipment is current.
|Equipment Code||Description||Start Date||End Date|
Event Log Summary
|Event Date||Event Type||Comments|
|11/6/2018 8:20:53 AM||General||On 5-30-2018 the CAN bus cabling was damaged taking all antennas offline. The CAN bus cables as well as IS1001 enclosures were rebuilt and installed on 10-31-2018 at 14:35. Antenna 01 is currently not connected.|
|2/28/2017 2:00:00 PM||General||At this time, power issues at LMC were restored. Connection to antenna 1 has been fixed and antennas 02 and 03 have been replaced with 25' X 8' flexible pass-through antennas. Initial results have shown a high performance of both the pass-through antennas with a complete field and ~24" of read range off the coil in all directions.|
|2/15/2016 11:00:00 AM||General||On 2/15/16, at around 11:00, antenna 3 became damaged due to high flows. By 15:30 antenna 2 was disconnected and was removed from sequence.|
|11/17/2015 9:00:00 PM||General||Around 9:00 PM antenna 3 lost connection. On 11/24/2015 at 2:04 PM, the connection problem was fixed, antenna 3 was tuned and resumed sequence.|
|3/6/2015 10:41:46 AM||General||Site has operated without any interruptions since it has been installed.|
|8/28/2014 8:18:39 AM||General||Registered new interrogation site LMC.|