Interrogation Site Metadata
Search by typing either a site name, site code, organization, or site type.
OKS - Shingle CreekStatus: Active - Operational
General Operational Period: Year Round
Start Date: 9/9/2016 12:00 AM
One 15-foot HDPE pipe antenna is installed in Shingle Creek, a tributary to the Okanagan River in Canada, immediately adjacent to the Okanagan Nation Alliance (ONA) fish Hatchery. The antenna is connected to an IS1001 and controlled by a Biomark Master Controller. The MC is housed in an enclosure mounted outside the hatchery. The site is approximately 1km from the creek mouth and uses AC power. The site was originally installed with four antennas and a QuBE controller, but was updated to current configuration in April 2019.
|Last File Loaded:||OKS-2021-183-P-01.xml|
|Date Last File Loaded:||8/30/2021 1:01 PM|
|Date Last File Opened:||7/2/2021 11:19 AM|
|Date Last File Closed:||8/30/2021 11:20 AM|
|Last Observation Tag:||3DD.0077CE54BC|
|Last Observation Date:||6/28/2021 11:32 PM|
|Detection Summary:||Last Day's Observations|
Data Steward:Skyeler Folks
Operational history describes the period the site was interrogating for PIT tags. The comments provide anecdotal information about the operation of the site during this period. Site operational periods are updated typically at the end of the calendar year.
|Year||Start Date||End Date||Comments|
|2020||1/1/2020 12:00 AM||12/9/2020 12:00 AM||Start and end dates based on timer tags. See event logs for additional operational information.|
|2019||9/5/2019 12:00 AM||12/31/2019 12:00 AM||Start and end dates based on timer tags.|
|2018||3/24/2018 12:00 AM||11/2/2018 12:00 AM|
|2017||4/5/2017 12:00 AM||9/14/2017 12:00 AM|
|2016||9/9/2016 12:00 AM||9/30/2016 12:00 AM|
|2015||No data received.|
Configuration metadata includes details about the equipment used at the site to detect and record PIT tags. A new configuration sequence record is created when equipment is added or removed from the site and the period of time it was active. A configuration sequence without an End Date is currently active.
Select a type to view the history of equipment installed at this site. A missing End Date implies the information about the equipment is current.
|Equipment Code||Description||Start Date||End Date|
Event Log Summary
|Event Date||Event Type||Comments|
|5/31/2021 3:29:39 PM||General||Site in Operation; Downloaded Pit Data|
|9/6/2019 11:00:00 AM||General||Swapped out antenna for a new antenna. Noise level dropped from over 80% to 50-55%.|
|7/9/2019 11:27:35 AM||SiteUp||87% FDBX noise!|
|4/11/2019 8:30:00 AM||General||Removed old nodes and antennas (two 10 footers from the stream). Installed new node, Master Controller, 15-foot antenna spanning the entire creek. Tested and tuned nodes, discovered high noise (90%), site not yet powered up until we solve the power/noise issue.|
|6/8/2018 3:45:00 PM||General||1 Antenna of 2 operational. Approximately 80% of width is being sampled. System sustained damage in spring freshet.|