Interrogation Site Metadata
Search by typing either a site name, site code, organization, or site type.
OMF - Omak Creek above Mission FallsStatus: Active - Non-Operational
General Operational Period: Year Round
Start Date: 3/3/2015 12:00 AM
OMF is a permanent PIT tag interrogation system located above Mission Falls in Omak Creek. Omak Creek enters the Okanogan River at RKM 51.5, approximately 1 KM upriver from the town of Omak, WA. The OMF site is located approximately 10.5 RKM upstream from the confluence of the Okanogan River. In March, 2015 the site consisted of a FS1001M running 4 pass-over PVC antennas, in two separate channel spanning rows. In March, 2018 the instream array was reconfigured with two 10 foot HDPE antennas anchored to the stream bed for pass-over detection. Antennas are arranged in a single row that spans the wetted width at base flow. Each antenna is power by a Biomark IS1001 reader multiplexed by an IS1001-MTS. Power is provided by a two 280 watt solar panels.
|Last File Loaded:||OMF-2021-277-P-001.xml|
|Date Last File Loaded:||11/1/2021 5:11 PM|
|Date Last File Opened:||10/4/2021 9:00 AM|
|Date Last File Closed:||11/1/2021 3:00 PM|
|Last Observation Tag:||3DD.0077C0F3B6|
|Last Observation Date:||10/28/2021 12:28 PM|
|Detection Summary:||Last Day's Observations|
Data Steward:Randy Johnson
Operational history describes the period the site was interrogating for PIT tags. The comments provide anecdotal information about the operation of the site during this period. Site operational periods are updated typically at the end of the calendar year.
|Year||Start Date||End Date||Comments|
|2020||3/4/2020 12:00 AM||11/16/2020 12:00 AM||Start and end dates based on timer tags. See event logs for additional operational information.|
|2019||3/19/2019 12:00 AM||6/27/2019 12:00 AM||Start and end dates based on timer tags.|
|2018||4/17/2018 12:00 AM||12/11/2018 12:00 AM||Start and end dates based on timer tags.|
|2017||3/23/2017 12:00 AM||10/18/2017 12:00 AM||Start and end dates based on timer tags.|
|2016||2/29/2016 12:00 AM||11/18/2016 12:00 AM||Start and end dates based on timer tags.|
|2015||3/3/2015 12:00 AM||11/24/2015 12:00 AM||Start and end dates based on timer tags.|
Configuration metadata includes details about the equipment used at the site to detect and record PIT tags. A new configuration sequence record is created when equipment is added or removed from the site and the period of time it was active. A configuration sequence without an End Date is currently active.
|Antenna Group||Transceiver ID||Antenna ID|
Select a type to view the history of equipment installed at this site. A missing End Date implies the information about the equipment is current.
|Equipment Code||Description||Start Date||End Date|
Event Log Summary
This displays the last 25 events, to see the full history, log in to the Dashboard and go to Interrogation Site > Event Logs.
|Event Date||Event Type||Comments|
|11/1/2021 5:00:00 PM||SiteDown||
Site shut down for season. Solar cannot keep up with demand.
|3/2/2021 10:00:00 AM||SiteUp||Site activated for season.|
|11/15/2020 8:29:00 AM||SiteDown||MC went to Standby due to low voltage. Site down for the rest of the season, solar cannot keep up.|
|11/10/2020 6:54:00 AM||General||MC went to Standby due to low voltage from 06:54 to 09:54 on 11/10/20, and then again from 17:54 until 09:22 on 11/14/20|
|11/9/2020 6:54:00 AM||General||MC went to Standby due to low voltage from 06:54 AM to 10:54 AM|
|9/30/2020 4:00:00 PM||SiteUp||Replaced canbus cable and power wires from solar panel to solar charge controller. Site back online.|
|9/8/2020 12:00:00 AM||SiteDown||Cold Springs Fire burned cabling from Solar Panel to Charge controller, and from MTS to ACN's. Site became inoperable.|
|3/4/2020 1:30:00 PM||SiteUp||Equipment reinstalled and site is back up and running.|
|7/18/2019 2:00:00 PM||SiteDown||Equipment was discovered stolen on July 18th. Last available data is from June 27th.|
|3/4/2019 12:00:00 PM||SiteUp||Site activated for the year. Solar is now able to keep up with power demands.|
|12/12/2018 3:48:00 PM||SiteDown||Site shut down for winter, solar could no longer meet power requirements.|
|3/27/2018 12:00:00 PM||General||Site reconfigured with one row of two 10 foot HDPE antennas anchored to the stream bed for pass-over detection.|
|3/15/2018 12:00:00 PM||SiteUp||Site activated, only antenna F4 is functional.|
|12/21/2017 12:00:00 PM||SiteDown||Site deactivated|
|3/27/2017 8:23:00 AM||SiteUp||Site activated, only antenna F4 is functional.|
|11/17/2016 11:00:00 AM||SiteDown||Download and removed system. Solar in 52.1v; out 22.6v.|
|11/9/2016 1:15:00 PM||General||Download. Solar in 61.1v; out 23.5v.|
|11/3/2016 1:00:00 PM||General||Download. Current good. Noise low. Solar in 62.2v; out 23.9v.|
|10/24/2016 7:00:00 AM||General||Download. Current good. Noise low. Solar in 64.2v; out 24.7v.|
|9/22/2016 10:30:00 AM||General||Download. All antennas 3-4 amps, 10-16% noise.|
|8/30/2016 7:30:00 AM||General||Download. All antennas functioning. Retuned all antennas.|
|8/25/2016 9:45:00 AM||General||Install antenna 4 and reset. 4.30 amps.|
|8/18/2016 9:30:00 AM||General||Antennas 1-3 reset and tuned. Antenna 4 not responsive. Needs replaced.|
|8/17/2016 8:30:00 AM||General||Download. Turn off system to work on antennas. 1230 antenna 1, 2 reset and retuned. Antenna 1 4.78amps 30% noise. Antenna 2 4.80amps 17% noise. Antennas 3, 4 need to be taped at new solder points and set in creek.|
|8/11/2016 8:00:00 AM||General||MUX unresponsive. Pulled coin cell and reset MUX settings. Took data from CF card. Antenna 1, 3 not working.|