Interrogation Site Metadata
SA1 - Salmon Creek Instream ArrayStatus: Active - Operational
General Operational Period: Year Round
Start Date: 2/8/2011 12:00 AM
Salmon Creek enters the Okanogan River at RKM 41.3, in the city of Okanogan, WA. The site was originally located 2.9 km upstream from the confluence with the Okanogan River, and originally consisted of 3 pass-over 7 foot PVC antennas powered by a FS1001M transceiver. Two of the antennas were damaged in spring/summer 2016 and subsequently removed. In February 2017, a second antenna was again added at the site. In January 2018, the site was moved approximately 1 km downstream and reinstalled with 4 new 20 ft HPDE antennas with an IS1001 master controller. Power is provided by Thermal Electric Generator (TEG).
|Last File Loaded:||SA1-2022-177-P-00A.xml|
|Date Last File Loaded:||6/26/2022 7:21 AM|
|Date Last File Opened:||6/26/2022 1:11 AM|
|Date Last File Closed:||6/26/2022 5:11 AM|
|Last Observation Tag:||3DD.003D98105C|
|Last Observation Date:||6/24/2022 10:41 PM|
|Detection Summary:||Last Day's Observations|
Data Steward:Randy Johnson
Operational history describes the period the site was interrogating for PIT tags. The comments provide anecdotal information about the operation of the site during this period. Site operational periods are updated typically at the end of the calendar year.
|Year||Start Date||End Date||Comments|
|2021||1/1/2021 12:00 AM||1/1/2022 12:00 AM||Start and end dates based on timer tags.|
|2020||1/1/2020 12:00 AM||1/1/2021 12:00 AM||Start and end dates based on timer tags. See event logs for additional operational information.|
|2019||1/1/2019 12:00 AM||12/31/2019 12:00 AM||Start and end dates based on timer tags.|
|2018||1/11/2018 12:00 AM||1/1/2019 12:00 AM||Site moved approximately 1km downstream and reinstalled with 4 new antennas and a IS1001 master controller. End date based on timer tags.|
|2017||1/19/2017 12:00 AM||5/26/2017 12:00 AM||Start and end dates based on timer tags.|
|2016||1/1/2016 12:00 AM||12/29/2016 12:00 AM||Antenna 3 down April 18. Antenna 1 down June 9. Both antennas removed in July. In August, original Mux replaced with unit that has only two functioning ports and the remaining antenna was reassigned ID B4. End date based on timer tags.|
|2015||1/1/2015 12:00 AM||1/1/2016 12:00 AM|
|2014||1/1/2014 12:00 AM||1/1/2015 12:00 AM|
|2013||1/1/2013 12:00 AM||1/1/2014 12:00 AM|
|2012||1/1/2012 12:00 AM||1/1/2013 12:00 AM|
|2011||2/8/2011 12:00 AM||1/1/2012 12:00 AM||First year of operations started on February 8. All three antennas became non-operational between May 20 and 23; site down. System reinstalled and operational on October 11. System down due to power isses from October 28 - November 22.|
Configuration metadata includes details about the equipment used at the site to detect and record PIT tags. A new configuration sequence record is created when equipment is added or removed from the site and the period of time it was active. A configuration sequence without an End Date is currently active.
|Antenna Group||Transceiver ID||Antenna ID|
Site Configuration Diagrams:
Select a type to view the history of equipment installed at this site. A missing End Date implies the information about the equipment is current.
|Equipment Code||Description||Start Date||End Date|
Event Log Summary
This displays the last 25 events, to see the full history, log in to the Dashboard and go to Interrogation Site > Event Logs.
|Event Date||Event Type||Comments|
|6/6/2022 8:12:00 AM||SiteUp||
Reset GFCI. Site has power again and system is functional.
|6/2/2022 7:06:00 PM||SiteDown||
Storm had tripped GFCI on outlet, shutting down power to the site.
|3/1/2022 10:53:00 AM||SiteUp||
GFCI switch reset. Site operating again.
|2/28/2022 10:30:00 AM||SiteDown||
GFCI switch had reset at power pole, shutting site down.
|6/1/2020 9:00:00 AM||SiteUp||Reset GFCI. Site up and running.|
|5/31/2020 6:00:00 AM||SiteDown||Power surge caused GFCI to trip. Site shut down.|
|10/29/2019 11:00:00 AM||General||Updated firmware on all equipment. Site had periods of intermittent data loss from 10/7 to 10/29/19.|
|10/7/2019 11:00:00 AM||SiteUp||Replaced MC and main CANBUS cable. Site back in operation and collecting data.|
|9/18/2019 11:00:00 AM||SiteDown||MC stopped working due to rodent chewing on main CANBUS cable.|
|4/16/2019 12:00:00 PM||SiteUp||Replaced MTS, site operating well.|
|4/12/2019 1:00:00 AM||SiteDown||Issue with MTS not powering on.|
|5/9/2018 11:00:00 PM||General||Readers for antennas 03, and 04 are no longer in communication with the MTS. As a result noise levels on antennas 01, and 02 are higher then normal. Expect lower detection efficiency from readers 01, and 02 till all 4 of the readers and antennas are back in communication with the MTS.|
|1/15/2018 7:42:00 AM||General||Installed Campbell Scientific CR1000, and PS100 to log interrogation and environmental data. A RavenX cellular modem was also installed to provide remote communication.|
|1/11/2018 7:42:00 AM||SiteUp||Installed four 20 foot HDPE antennas anchored to the stream bed for pass-over detection. Each antenna is power by a IS1001 reader housed in a canister connected to each antenna. All readers are multiplexed through a single canbus cable connected to a newly installed IS1001-MC.|
|6/28/2017 10:23:46 AM||SiteDown||Per Brian Miller, the antennas were blown out during the high flows and the site has been shut down until further notice.|
|12/7/2016 11:15:00 AM||General||Swap out low voltage batteries. Reduce air/gas PSI to 7.5.|
|12/6/2016 9:00:00 AM||General||Low voltage ~20v on TEG. Increased PSI from 7.8 to 8.5. Battery 11.3v and 8.4v. Need changed.|
|10/24/2016 7:30:00 AM||General||Antenna 2 4.6 amps, 17% noise.|
|10/24/2016 7:30:00 AM||General||Antenna2 4.6amps 17% noise|
|8/30/2016 3:30:00 PM||General||Reconnect MUX from Omak Creek (OMK) at SA1 site. Using MUX port 4 connected to antenna2 (only port 4 and port 6 work properly). Retuned at 4.4amps. Note that B4 will be represented by antenna2. The only functioning antenna at the site.|
|8/30/2016 3:30:00 PM||SiteUp||Reconnect OMK (Lower Omak Creek) MUX using MUX port 4 for antenna 2. All other antennas (1 and 3) were destroyed. Retuned to 4.4 amps, 60% noise.|
|8/30/2016 1:30:00 PM||SiteDown||Remove MUX from site and take to OMK (Lower Omak Creek).|
|8/30/2016 1:30:00 PM||General||Remove MUX from site and move to Omak Creek (OMK) to replace limited functioning MUX.|
|7/26/2016 1:30:00 PM||General||Remove antenna1 and antenna3. Both antennas not functioning. Antenna1 no current, frayed pigtail. Antenna3 no current, cracked. Antenna1 went down ~6/9/16 16:00. Antenna3 went down ~4/18/16 14:00.|
|7/26/2016 1:30:00 PM||General||Remove antennas 1, 3. Antenna 1 no current, frayed pigtail. Antenna 3 no current, cracked. TEG turned off, turned back on ~24v.|