Interrogation Site Metadata
Search by typing either a site name, site code, organization, or site type.
LEA - Leaburg Dam Combined FishwaysStatus: Decommissioned
General Operational Period: Year Round
Start Date: 12/1/2009 12:00 AM
End Date: 9/5/2018 12:00 AM
Leaburg Dam is located at km 56 on the McKenzie River, east of Eugene, OR. The screened bypass diverts downstream migrants from the turbine intakes. Dual FDX/HDX PIT tag detectors were installed in the bypass exit in December 2009. Smolts can also pass through the spillway. There is a fish ladder located on each bank. The antennas in the fish ladders never reported data to PTAGIS. Site was upgraded from RM310 reader boards to IS1001s in May 2016. The IS1001s were set to read FDX tags only. A third antenna was added June 2017. Site is no longer active.
Primary Contact:Luke Whitman
Data Steward:Derrek Faber
Operational history describes the period the site was interrogating for PIT tags. The comments provide anecdotal information about the operation of the site during this period. Site operational periods are updated typically at the end of the calendar year.
|Year||Start Date||End Date||Comments|
|2018||1/1/2018 12:00 AM||9/5/2018 12:00 AM||Start and end dates based on timer tags.|
|2017||1/1/2017 12:00 AM||1/1/2018 12:00 AM||Start and end dates based on timer tags.|
|2016||1/1/2016 12:00 AM||1/1/2017 12:00 AM||Start and end dates based on timer tags.|
|2015||1/1/2015 12:00 AM||1/1/2016 12:00 AM|
|2014||1/1/2014 12:00 AM||1/1/2015 12:00 AM|
|2013||1/1/2013 12:00 AM||1/1/2014 12:00 AM|
|2012||1/1/2012 12:00 AM||1/1/2013 12:00 AM|
|2011||1/1/2011 12:00 AM||1/1/2012 12:00 AM||Site down from October 4-6 for antenna installations. No data from October 24-November 2; readers were inadvertantly switched off.|
|2010||1/1/2010 12:00 AM||1/1/2011 12:00 AM|
|2009||12/1/2009 12:00 AM||1/1/2010 12:00 AM||First year of operation.|
Configuration metadata includes details about the equipment used at the site to detect and record PIT tags. A new configuration sequence record is created when equipment is added or removed from the site and the period of time it was active. A configuration sequence without an End Date is currently active.
|Antenna Group||Transceiver ID||Antenna ID|
|Screened Smolt Bypass Channel||01||01|
|Screened Smolt Bypass Channel||02||02|
|Screened Smolt Bypass Channel||03||03|
Select a type to view the history of equipment installed at this site. A missing End Date implies the information about the equipment is current.
|Equipment Code||Description||Start Date||End Date|
Event Log Summary
|Event Date||Event Type||Comments|
|6/29/2017 4:00:00 PM||SiteUp||Three operational antennas with site back up and running. Water returned to bypass canal.|
|6/29/2017 9:55:00 AM||SiteDown||Replaced three antennas with FRP-clad, ferrite shielded antennas. All three antennas interrogate entire 5'x5' channel, and are synced to each-other through IS-1001 24V readers. Previous setup had two operational antennas, now the system has three operational antennas. Moved power to a dedicated breaker, on battery backup.|
|6/28/2017 9:55:00 AM||SiteDown||Site down for antenna upgrade.|
|3/25/2017 12:00:00 AM||SiteDown||EWEB conducted annual maintenance on Leaburg canal from 3/25/17 to 4/25/17. No detections during this period; fish bypass dewatered.|
|5/4/2016 1:00:00 PM||SiteUp||Site operational.|
|5/4/2016 12:02:00 PM||General||Leaburg site upgraded from RM-310 reader boards to two IS1001 reader boards. Antennas retuned and data collection software installed on cabinet computer. System was tested with 10 live fish with HDX on and only 7 were detected. HDX was turned off and the system was tested with 20 fish and all 20 were detected. HDX was left off. Currently, the system is operating with two antennas, separated by 25’. Antenna one is the upstream antenna, and antenna two is the downstream antenna. These antennas provide full coverage of the ~5’x5’ fish return at the Leaburg canal fish screen.|
|5/3/2016 9:20:00 AM||SiteDown||Site shut down for reader upgrade from RM-310 to IS1001.|
|5/13/2014 12:01:00 AM||SiteUp||Ladder operations began along with PIT tag monitoring.|