Interrogation Site Metadata
Search by typing either a site name, site code, organization, or site type.
NMC - Ninemile Creek Instream ArrayStatus: Active - Operational
General Operational Period: Year Round
Start Date: 2/12/2011 11:09 AM
Ninemile enters east side of Lake Osoyoos at Okanogan River RKM 129.5, north of the town of Oroville, WA. Site NMC is located on Ninemile Creek, 0.78 km upstream from the confluence with Lake Osoyoos. The setup consists of 3 pass-through HDPE antennas (a fourth original antenna was removed September 2019) accompanied by individual IS-1001 reader boards placed in series. All antennas are 10 feet in length, with antenna 1 the most upstream, antenna 2 the middle, and antenna 3 the downstream. The site is on grid power and regulated by an Acopian linear power supply to an MTS-MC. A bioprobe collects and stores both air and water temperatures as well as the water depth data. The NMC site has a Cloudgate 4G cellular modem which transmits data to PTAGIS through Biomark.
|Last File Loaded:||NMC-2021-295-P-00D.xml|
|Date Last File Loaded:||10/22/2021 7:40 PM|
|Date Last File Opened:||10/22/2021 2:23 PM|
|Date Last File Closed:||10/22/2021 6:23 PM|
|Last Observation Tag:||3DD.003D372460|
|Last Observation Date:||10/22/2021 6:23 PM|
|Detection Summary:||Last Day's Observations|
Data Steward:Randy Johnson
Operational history describes the period the site was interrogating for PIT tags. The comments provide anecdotal information about the operation of the site during this period. Site operational periods are updated typically at the end of the calendar year.
|Year||Start Date||End Date||Comments|
|2020||1/1/2020 12:00 AM||1/1/2021 12:00 AM||Start and end dates based on timer tags. See event logs for additional operational information.|
|2019||1/1/2019 12:00 AM||12/31/2019 12:00 AM||Start and end dates based on timer tags.|
|2018||1/1/2018 12:00 AM||1/1/2019 12:00 AM||Start and end dates based on timer tags.|
|2017||1/1/2017 12:00 AM||1/1/2018 12:00 AM||Start and end dates based on timer tags.|
|2016||1/1/2016 12:00 AM||1/1/2017 12:00 AM||Start and end dates based on timer tags.|
|2015||1/1/2015 12:00 AM||1/1/2016 12:00 AM|
|2014||1/1/2014 12:00 AM||1/1/2015 12:00 AM|
|2013||1/1/2013 12:00 AM||1/1/2014 12:00 AM|
|2012||1/1/2012 12:00 AM||1/1/2013 12:00 AM|
|2011||2/12/2011 11:09 AM||1/1/2012 12:00 AM||First year of operations.|
Configuration metadata includes details about the equipment used at the site to detect and record PIT tags. A new configuration sequence record is created when equipment is added or removed from the site and the period of time it was active. A configuration sequence without an End Date is currently active.
Select a type to view the history of equipment installed at this site. A missing End Date implies the information about the equipment is current.
|Equipment Code||Description||Start Date||End Date|
Event Log Summary
|Event Date||Event Type||Comments|
|10/26/2020 4:00:00 PM||SiteUp||Replaced MTS-MC and Acopian Linear Power supply, removed the Bioprobe board, and added a circuit breaker between Acopian and terminal block.|
|10/23/2020 10:00:00 AM||SiteDown||A short caused multiple equipment failures, possibly by a stink bug.|
|9/2/2019 12:30:00 PM||General||Upgraded from a MUX to an IS-1001 MTS-MC and installed 3 new IS-1001 reader boards. Removed the 4 old PVC antennas, drove new T-posts and replaced with 3 new HDPE antennas in pass-through orientation.|
|3/9/2016 11:15:00 AM||General||Modem not responding. No network or signal. Modem power moved from 12v to sw12v and reset. Showing network and signal.|
|11/15/2014 1:00:00 AM||General||Unknown file gap.|
|5/23/2013 1:00:00 PM||General||Download CR1000 and install Biomark 8.0 software|
|1/15/2013 2:00:00 PM||General||Replace battery bank A due to low voltage. Batteries at 7v. Restart system and tune antennas @ 1615|