Interrogation Site Metadata
Search by typing either a site name, site code, organization, or site type.
WRU - Upper Wind River (WA) rkm 30Status: Active - Operations Unspecified
General Operational Period: Year Round
Start Date: 10/12/2012 12:00 AM
An instream interrogation system on the Wind River, WA. In its original configuration the site was located at river kilometer 30 near the FR3065 bridge over the Wind River and had 6 antennas in two arrays. After a flood event in December 2015 damaged all but one of the antennas, the site operated sporadically until it was relocated in October 2016 to a location 2 km downstream of its original location and reinstalled with three 20ft antennas. In September 2017, a 2nd array of three antennas was added to the site.
|Last File Loaded:||WRU-2021-293-P-A.xml|
|Date Last File Loaded:||10/20/2021 3:01 PM|
|Date Last File Opened:||10/14/2021 9:38 AM|
|Date Last File Closed:||10/20/2021 10:43 AM|
|Last Observation Tag:||3DD.003D6F9A9D|
|Last Observation Date:||10/15/2021 6:46 PM|
|Detection Summary:||Last Day's Observations|
Data Steward:Ian Jezorek
Operational history describes the period the site was interrogating for PIT tags. The comments provide anecdotal information about the operation of the site during this period. Site operational periods are updated typically at the end of the calendar year.
|Year||Start Date||End Date||Comments|
|2020||1/1/2020 12:00 AM||1/1/2021 12:00 AM||Start and end dates based on timer tags. See event logs for additional operational information.|
|2019||1/1/2019 12:00 AM||12/31/2019 12:00 AM||Start and end dates based on timer tags.|
|2018||1/1/2018 12:00 AM||1/1/2019 12:00 AM||Antennas 2 and 3 replaced July 11, all antennas now operational.|
|2017||1/1/2017 12:00 AM||1/1/2018 12:00 AM||Second antenna array of 3 antennas added September 8. Antenna 3 lost connection November 23. Antenna 2 lost connection December 29.|
|2016||5/23/2016 12:00 AM||1/1/2017 12:00 AM||Operated sporadically during spring with only one antenna. Moved 2 km downstream in October and reinstalled with 3 antennas|
|2015||1/1/2015 12:00 AM||12/9/2015 12:00 AM||Antenna 2 down or operating poorly January 27-March 3. Flood event shut site down and damaged all antennas except for F3. Flood event in Decemeber damaged all antennas except for F3 and site was shut down.|
|2014||1/1/2014 12:00 AM||1/1/2015 12:00 AM||Site down for 2.5 hours on February 28. Site down from March 4-7. Antenna 1 down from March 7-June 23.|
|2013||1/1/2013 12:00 AM||1/1/2014 12:00 AM||Site down multiple times in May due to power issues, see event log for details. Antenna 1 and 2 down until September 3. Antenna 2 down September 29-October 22. Site down November 16-19.|
|2012||10/12/2012 12:00 AM||1/1/2013 12:00 AM|
Configuration metadata includes details about the equipment used at the site to detect and record PIT tags. A new configuration sequence record is created when equipment is added or removed from the site and the period of time it was active. A configuration sequence without an End Date is currently active.
|Antenna Group||Transceiver ID||Antenna ID|
Select a type to view the history of equipment installed at this site. A missing End Date implies the information about the equipment is current.
|Equipment Code||Description||Start Date||End Date|
Event Log Summary
This displays the last 25 events, to see the full history, log in to the Dashboard and go to Interrogation Site > Event Logs.
|Event Date||Event Type||Comments|
|9/7/2021 3:37:00 PM||General||
Replaced antenna 1 with a new HDPE antenna. Back to running a full 6 antenna system.
|4/8/2021 4:35:29 PM||General||On 4/8/2021 at about 1300 we connected a new cable to Antenna 3 to replace the one that was cut during the Jan 2021 flood. Ian|
|1/20/2021 8:39:24 AM||General||Early morning of 13 January 2021 high flow and debris broke antenna 1 and severed the cable to antenna 3. We will repair and replace asap, but will require lower flow. Ian|
|7/23/2018 11:50:32 AM||General||On 7/11/2018 antennas 2 and 3 at WRU, which had been broken the previous winter, were replaced. Site now had six functioning antennas. Ian|
|1/2/2018 4:31:04 PM||General||At about midnight on December 29, 2017, Antenna 2 at WRU was broken. Potentially by the large log still hung up on Antenna 3. Both antennas are out of sequence. Ian|
|11/29/2017 2:55:01 PM||General||Around 7am on Nov 23rd a large tree/rootwad washed up near antenna 3 and we lost contact with antenna 3. Do not know at this time if the antenna is damaged or the cable has just been disconnected. Antenna 3 is currently out of the antenna sequence. Ian|
|9/11/2017 11:55:34 AM||General||A second antenna array was added to this site on 9/8/2017. Upstream array is Antennas 1, 2, 3; downstream array is antennas 4, 5, 6. Ian|
|11/3/2016 1:18:17 PM||General||Site WRU was reactivated on 6 Oct 2016. The site was relocated about 1.5km downstream of its former location, where it was destroyed in the December 2015 flood. We hope the new location will be friendlier at high water. Current configuration is one array of three 20' antennas. We hope to add a second array during 2017. Ian|
|6/8/2016 1:27:30 PM||General||The flood of Dec 9 2015 destroyed all but antenna 3 at this site. Our MUX unit was malfunctioning during the winter so the site was offline for awhile. On 23 May 2016 we installed a new MUX and began operating just antenna 3. We will replace antennas during the summer of 2016. Ian|
|12/9/2015 4:45:54 PM||General||During the night of 12/8, contact was lost with Antennas 2, 4, 5, and 6. Major flood, unknown if pulled cables or damaged antennas. Ian|
|12/8/2015 4:11:39 PM||General||At about 1500hrs on 12/7/15, contact was lost with Antenna 1. Will have to wait for low water, hopefully pulled cable. Ian|
|12/7/2015 3:06:39 PM||General||On 12/3/2015 at about 12:30, the number one antenna at WRU was reconnected. Ian|
|11/24/2015 5:08:02 PM||General||On the night of November 17 2015, flooding disconnected antenna 1. Will have to wait for flow to drop to get in and reconnect the antenna. Ian|
|6/8/2015 2:08:16 PM||General||A short in the charging circuit at WRU (possibly caused by ants) caused loss of power to the MUX from about 2100 on 6/6/2015 to about 1000 on 6/8/2015. Ian|
|5/26/2015 4:24:55 PM||General||WRU lost power and due to a bad lithium battery also data from 19 May at 12:00 to 25 May at 11:00. Lithium battery was replaced on 26 May. Ian|
|5/20/2015 2:23:57 PM||General||WRU lost power from about 12:45 on 18 May to about 0900 on 19 May 2015. Also from about 12:00 19 May to about 08:45 on 20 May. Disruptions have been due to poor solar generation. Panels have been moved and batteries recharged to hopefully ameliorate the problem. Ian|
|5/19/2015 8:41:25 AM||General||WRU lost and regained power several times between 5/14/15 and 5/18/15. Correct time and dates were lost. Interrogation times during this period are estimates.|
|3/4/2015 4:38:58 PM||General||On 27 January 2015, the cable to antenna two developed a frayed spot due to friction on a rock. This resulted in noise to the antenna that prevented tag reads. The cable was replaced on 3 March 2015 and antenna two is back to normal operation.|
|2/11/2015 3:18:00 PM||General||WRU was down due to power loss from 0200 to 1100 on 11 February 2015|
|10/28/2014 3:18:00 PM||General||WRU was down due to power loss from 28 October at 10:00 to 31 October at 13:00.|
|10/24/2014 3:18:00 PM||General||WRU was down due to power loss on 24 October 2014|
|6/23/2014 3:00:00 PM||General||The cable to antenna number one was reconnected on 23 June 2014. Antenna one is now back in sequence. Ian|
|3/11/2014 9:14:48 AM||General||Power was lost to WRU from March 4 2014 at 19:30 to March 7 2014 at 12:12. Also, during this period, contact was lost with Antenna 1. Water has been very high recently. Cannot see if there is antenna damage or just a pulled cable. We will investigate once conditions allow.|
|3/3/2014 9:48:16 AM||General||On February 28, the site was down from about 10:30 to 1pm for maintenance and battery charging. Ian|
|11/19/2013 4:57:44 PM||General||Power was lost at WRU from 1900 hrs on 11/16/2013 to 1600 hrs on 11/19/2013. Ian|